3-Phase 4-Wire Power Quality Analyzer Compliant to IEC61000-4-30 Class S Standards and On-screen Guide to Deliver Easy Setup and Testing
Basic specifications (Accuracy guaranteed for 1 year, Post-adjustment accuracy guaranteed for 1 year)
Measurement line type | Single-phase 2-wire, Single-phase 3-wire, Three-phase 3-wire or Three-phase 4-wire plus one extra input channel CH4 for voltage/current, (all channels AC/DC measurement) |
---|---|
Voltage ranges | Voltage measurement: 1000.0 V rms or DC, Transient measurement 2.200 kV peak |
Current ranges |
50.000 mA AC to 5.0000 kA AC, 10.000 A DC to 2.0000 kA DC (depends on current sensor in use) |
Power ranges | 50.000 W to 6.0000 MW (determined automatically based on current range in use) |
Basic accuracy | Voltage: ±0.2% of nominal voltage, Current: ±0.1 % rdg. ±0.1 % f.s. + current sensor accuracy, Active power: DC ±0.5 % rdg. ±0.5 % f.s. + current sensor accuracy, AC ±0.2 % rdg. ±0.1 % f.s. + current sensor accuracy |
Measurement items |
1. Transient over voltage : 200 kHz sampling 2. Frequency cycle : Calculated as one cycle 3. Voltage (1/2) RMS, Current (1/2) RMS: one cycle calculation refreshed every half cycle 4. Voltage swell, Voltage dips, Voltage interruption 5. Inrush current : half-cycle calculation 6. Frequency 200 ms: Calculated as 10 or 12 cycles 7. 10-sec frequency: Calculated as the whole-cycle time during the specified 10 s period 8. Voltage waveform peak, Current waveform peak 9. Voltage, Current, Active power, Apparent power, Reactive power, Active energy, Apparent energy, Reactive energy, Energy cost, Power factor, Displacement power factor, Voltage unbalance factor, Current unbalance factor 10. Voltage crest factor, Current crest factor 11. Harmonic/ Harmonic phase angle (voltage/ current), Harmonic power: 0 th to 50 th orders 12. Harmonic voltage-current phase angle: 1 th to 50 th orders 13. Total harmonic distortion factor (voltage/ current) 14. Inter harmonic (voltage/ current): 0.5 th to 49.5 th orders 15. K Factor (multiplication factor) 16. (Available in future firmware update) Instantaneous flicker value: As per IEC61000-4-15 17. (Available in future firmware update) IEC Flicker, Δ V10 Flicker` |
Record | Maximum recording interval: 1 year, Maximum number of recordable events: 9999 × 365 days |
Interfaces | SD/SDHC card, RS-232C, LAN (HTTP server), USB 2.0 |
Display | 6.5-inch TFT color LCD (640 × 480 dots) |
Power supply | AC adapter Z1002 (100 V to 240 V AC, 50/60 Hz, rated current 1.7 A), Battery pack Z1003 (Continuous use: 8 hr, Charging time: Max. 5 hr 30 m with AC adapter) |
Dimensions and mass | 300 mm (11.81 in)W × 211 mm (8.31 in)H × 68 mm (2.68 in)D, 2.5 kg (88.2 oz) (including battery pack) |
Accessories |
Basic specifications (Accuracy guaranteed for 1 year, Post-adjustment accuracy guaranteed for 1 year)
Measurement line type | Single-phase 2-wire, Single-phase 3-wire, Three-phase 3-wire or Three-phase 4-wire plus one extra input channel CH4 for voltage/current, (all channels AC/DC measurement) |
---|---|
Voltage ranges | Voltage measurement: 1000.0 V rms or DC, Transient measurement 2.200 kV peak |
Current ranges |
50.000 mA AC to 5.0000 kA AC, 10.000 A DC to 2.0000 kA DC (depends on current sensor in use) |
Power ranges | 50.000 W to 6.0000 MW (determined automatically based on current range in use) |
Basic accuracy | Voltage: ±0.2% of nominal voltage, Current: ±0.1 % rdg. ±0.1 % f.s. + current sensor accuracy, Active power: DC ±0.5 % rdg. ±0.5 % f.s. + current sensor accuracy, AC ±0.2 % rdg. ±0.1 % f.s. + current sensor accuracy |
Measurement items |
1. Transient over voltage : 200 kHz sampling 2. Frequency cycle : Calculated as one cycle 3. Voltage (1/2) RMS, Current (1/2) RMS: one cycle calculation refreshed every half cycle 4. Voltage swell, Voltage dips, Voltage interruption 5. Inrush current : half-cycle calculation 6. Frequency 200 ms: Calculated as 10 or 12 cycles 7. 10-sec frequency: Calculated as the whole-cycle time during the specified 10 s period 8. Voltage waveform peak, Current waveform peak 9. Voltage, Current, Active power, Apparent power, Reactive power, Active energy, Apparent energy, Reactive energy, Energy cost, Power factor, Displacement power factor, Voltage unbalance factor, Current unbalance factor 10. Voltage crest factor, Current crest factor 11. Harmonic/ Harmonic phase angle (voltage/ current), Harmonic power: 0 th to 50 th orders 12. Harmonic voltage-current phase angle: 1 th to 50 th orders 13. Total harmonic distortion factor (voltage/ current) 14. Inter harmonic (voltage/ current): 0.5 th to 49.5 th orders 15. K Factor (multiplication factor) 16. (Available in future firmware update) Instantaneous flicker value: As per IEC61000-4-15 17. (Available in future firmware update) IEC Flicker, Δ V10 Flicker` |
Record | Maximum recording interval: 1 year, Maximum number of recordable events: 9999 × 365 days |
Interfaces | SD/SDHC card, RS-232C, LAN (HTTP server), USB 2.0 |
Display | 6.5-inch TFT color LCD (640 × 480 dots) |
Power supply | AC adapter Z1002 (100 V to 240 V AC, 50/60 Hz, rated current 1.7 A), Battery pack Z1003 (Continuous use: 8 hr, Charging time: Max. 5 hr 30 m with AC adapter) |
Dimensions and mass | 300 mm (11.81 in)W × 211 mm (8.31 in)H × 68 mm (2.68 in)D, 2.5 kg (88.2 oz) (including battery pack) |
Accessories |
Basic specifications (Accuracy guaranteed for 1 year, Post-adjustment accuracy guaranteed for 1 year)
Measurement line type | Single-phase 2-wire, Single-phase 3-wire, Three-phase 3-wire or Three-phase 4-wire plus one extra input channel CH4 for voltage/current, (all channels AC/DC measurement) |
---|---|
Voltage ranges | Voltage measurement: 1000.0 V rms or DC, Transient measurement 2.200 kV peak |
Current ranges |
50.000 mA AC to 5.0000 kA AC, 10.000 A DC to 2.0000 kA DC (depends on current sensor in use) |
Power ranges | 50.000 W to 6.0000 MW (determined automatically based on current range in use) |
Basic accuracy | Voltage: ±0.2% of nominal voltage, Current: ±0.1 % rdg. ±0.1 % f.s. + current sensor accuracy, Active power: DC ±0.5 % rdg. ±0.5 % f.s. + current sensor accuracy, AC ±0.2 % rdg. ±0.1 % f.s. + current sensor accuracy |
Measurement items |
1. Transient over voltage : 200 kHz sampling 2. Frequency cycle : Calculated as one cycle 3. Voltage (1/2) RMS, Current (1/2) RMS: one cycle calculation refreshed every half cycle 4. Voltage swell, Voltage dips, Voltage interruption 5. Inrush current : half-cycle calculation 6. Frequency 200 ms: Calculated as 10 or 12 cycles 7. 10-sec frequency: Calculated as the whole-cycle time during the specified 10 s period 8. Voltage waveform peak, Current waveform peak 9. Voltage, Current, Active power, Apparent power, Reactive power, Active energy, Apparent energy, Reactive energy, Energy cost, Power factor, Displacement power factor, Voltage unbalance factor, Current unbalance factor 10. Voltage crest factor, Current crest factor 11. Harmonic/ Harmonic phase angle (voltage/ current), Harmonic power: 0 th to 50 th orders 12. Harmonic voltage-current phase angle: 1 th to 50 th orders 13. Total harmonic distortion factor (voltage/ current) 14. Inter harmonic (voltage/ current): 0.5 th to 49.5 th orders 15. K Factor (multiplication factor) 16. (Available in future firmware update) Instantaneous flicker value: As per IEC61000-4-15 17. (Available in future firmware update) IEC Flicker, Δ V10 Flicker` |
Record | Maximum recording interval: 1 year, Maximum number of recordable events: 9999 × 365 days |
Interfaces | SD/SDHC card, RS-232C, LAN (HTTP server), USB 2.0 |
Display | 6.5-inch TFT color LCD (640 × 480 dots) |
Power supply | AC adapter Z1002 (100 V to 240 V AC, 50/60 Hz, rated current 1.7 A), Battery pack Z1003 (Continuous use: 8 hr, Charging time: Max. 5 hr 30 m with AC adapter) |
Dimensions and mass | 300 mm (11.81 in)W × 211 mm (8.31 in)H × 68 mm (2.68 in)D, 2.5 kg (88.2 oz) (including battery pack) |
Accessories |
Basic specifications (Accuracy guaranteed for 1 year, Post-adjustment accuracy guaranteed for 1 year)
Measurement line type | Single-phase 2-wire, Single-phase 3-wire, Three-phase 3-wire or Three-phase 4-wire plus one extra input channel CH4 for voltage/current, (all channels AC/DC measurement) |
---|---|
Voltage ranges | Voltage measurement: 1000.0 V rms or DC, Transient measurement 2.200 kV peak |
Current ranges |
50.000 mA AC to 5.0000 kA AC, 10.000 A DC to 2.0000 kA DC (depends on current sensor in use) |
Power ranges | 50.000 W to 6.0000 MW (determined automatically based on current range in use) |
Basic accuracy | Voltage: ±0.2% of nominal voltage, Current: ±0.1 % rdg. ±0.1 % f.s. + current sensor accuracy, Active power: DC ±0.5 % rdg. ±0.5 % f.s. + current sensor accuracy, AC ±0.2 % rdg. ±0.1 % f.s. + current sensor accuracy |
Measurement items |
1. Transient over voltage : 200 kHz sampling 2. Frequency cycle : Calculated as one cycle 3. Voltage (1/2) RMS, Current (1/2) RMS: one cycle calculation refreshed every half cycle 4. Voltage swell, Voltage dips, Voltage interruption 5. Inrush current : half-cycle calculation 6. Frequency 200 ms: Calculated as 10 or 12 cycles 7. 10-sec frequency: Calculated as the whole-cycle time during the specified 10 s period 8. Voltage waveform peak, Current waveform peak 9. Voltage, Current, Active power, Apparent power, Reactive power, Active energy, Apparent energy, Reactive energy, Energy cost, Power factor, Displacement power factor, Voltage unbalance factor, Current unbalance factor 10. Voltage crest factor, Current crest factor 11. Harmonic/ Harmonic phase angle (voltage/ current), Harmonic power: 0 th to 50 th orders 12. Harmonic voltage-current phase angle: 1 th to 50 th orders 13. Total harmonic distortion factor (voltage/ current) 14. Inter harmonic (voltage/ current): 0.5 th to 49.5 th orders 15. K Factor (multiplication factor) 16. (Available in future firmware update) Instantaneous flicker value: As per IEC61000-4-15 17. (Available in future firmware update) IEC Flicker, Δ V10 Flicker` |
Record | Maximum recording interval: 1 year, Maximum number of recordable events: 9999 × 365 days |
Interfaces | SD/SDHC card, RS-232C, LAN (HTTP server), USB 2.0 |
Display | 6.5-inch TFT color LCD (640 × 480 dots) |
Power supply | AC adapter Z1002 (100 V to 240 V AC, 50/60 Hz, rated current 1.7 A), Battery pack Z1003 (Continuous use: 8 hr, Charging time: Max. 5 hr 30 m with AC adapter) |
Dimensions and mass | 300 mm (11.81 in)W × 211 mm (8.31 in)H × 68 mm (2.68 in)D, 2.5 kg (88.2 oz) (including battery pack) |
Accessories |
Basic specifications (Accuracy guaranteed for 1 year, Post-adjustment accuracy guaranteed for 1 year)
Measurement line type | Single-phase 2-wire, Single-phase 3-wire, Three-phase 3-wire or Three-phase 4-wire plus one extra input channel CH4 for voltage/current, (all channels AC/DC measurement) |
---|---|
Voltage ranges | Voltage measurement: 1000.0 V rms or DC, Transient measurement 2.200 kV peak |
Current ranges |
50.000 mA AC to 5.0000 kA AC, 10.000 A DC to 2.0000 kA DC (depends on current sensor in use) |
Power ranges | 50.000 W to 6.0000 MW (determined automatically based on current range in use) |
Basic accuracy | Voltage: ±0.2% of nominal voltage, Current: ±0.1 % rdg. ±0.1 % f.s. + current sensor accuracy, Active power: DC ±0.5 % rdg. ±0.5 % f.s. + current sensor accuracy, AC ±0.2 % rdg. ±0.1 % f.s. + current sensor accuracy |
Measurement items |
1. Transient over voltage : 200 kHz sampling 2. Frequency cycle : Calculated as one cycle 3. Voltage (1/2) RMS, Current (1/2) RMS: one cycle calculation refreshed every half cycle 4. Voltage swell, Voltage dips, Voltage interruption 5. Inrush current : half-cycle calculation 6. Frequency 200 ms: Calculated as 10 or 12 cycles 7. 10-sec frequency: Calculated as the whole-cycle time during the specified 10 s period 8. Voltage waveform peak, Current waveform peak 9. Voltage, Current, Active power, Apparent power, Reactive power, Active energy, Apparent energy, Reactive energy, Energy cost, Power factor, Displacement power factor, Voltage unbalance factor, Current unbalance factor 10. Voltage crest factor, Current crest factor 11. Harmonic/ Harmonic phase angle (voltage/ current), Harmonic power: 0 th to 50 th orders 12. Harmonic voltage-current phase angle: 1 th to 50 th orders 13. Total harmonic distortion factor (voltage/ current) 14. Inter harmonic (voltage/ current): 0.5 th to 49.5 th orders 15. K Factor (multiplication factor) 16. (Available in future firmware update) Instantaneous flicker value: As per IEC61000-4-15 17. (Available in future firmware update) IEC Flicker, Δ V10 Flicker` |
Record | Maximum recording interval: 1 year, Maximum number of recordable events: 9999 × 365 days |
Interfaces | SD/SDHC card, RS-232C, LAN (HTTP server), USB 2.0 |
Display |
Back
Related Product Category Power Meters, Power Analyzers, Current Sensors
Copyright © 2014 Protindo Pratama - All rights reserved. Design & Powered by
LaWaveDesign.com -
Disclaimer
Best Viewed in Mozilla Firefox (3.0.1) or Greater |